I’ve been having problems with Carelink uploading. It only appears to work about 1 out of 5 times. The pump has to be very close to the Carelink Next glucose meter, and I have never been able to get this to work with the included USB extension cable.
I called Medtronics, and they suggested radio interference might be causing the problem, and suggest putting an aluminum turkey pan over the pump and meter to act as a shield; Real Professional.
I feel, if Medtronics designs such a flakey system, the least they could do is to included a shielded enclosure to place the pump and meter into while uploading. I’ve always gotten it to work, but it does require about 5 minutes of repositioning the pump and retrying every time.
Any other 670G users out there with this issue? I don’t think my unit is defective, I think the design is crap.